Please note we're taking an extra day over the Melbourne Cup weekend (i.e. we'll be closed Sat 30th Oct. through to Tues 3rd Nov.)
Lockdown rules DO still apply to retail. Please read our latest COVID info. (I.e. we're open for click and collect/mail order only).
This is the Image Science guide to getting the best out of your BenQ P Series monitor through BenQ's direct hardware calibration system, called Palette Master. We take you through the process from front to back and explain all the settings along the way.
Of course if you have any questions after you've reviewed this guide, don't hesitate to get in touch for extra advice.
(N.B. This guide is only for BenQ P series monitors (PV and PG models) - if you monitor's model number instead starts with an SW instead, follow this guide instead)
Before we start the actual calibration, we need to make sure all aspects of our system are Ready To Go.
Please first follow our guide to getting ready for monitor calibration - once you're done there, come back here for the rest (and don't skip this bit as this covers the areas where most folks run into issues!).
Once you have made the USB connection to the monitor, then connect your calibrator.
The most reliable approach is to now connect your calibrator to one of the downstream USB ports on the monitor itself. (Sometimes the BenQ software can be fussy about recognising the calibrator if you instead have it attached to a USB port on your computer - so if you have any issues in this area, try the ports on your monitor).
Start by plugging your calibrator in. Palette Master is compatible with, and comes with drivers for, X-Rite i1Display and i1Pro calibrators. (If you don't have a calibrator yet, we strongly recommend the X-Rite i1Display Pro - it's the de facto industry standard calibrator and an excellent, reliable, accurate device).
Make sure your monitor has properly warmed up before you start calibrating - with BenQ monitors just five minutes of switched on time is enough.
When Palette Master first opens, you will be greeted with an initial start-up screen. This screen should correctly identify your monitor and the calibrator you are using. If you forgot to plug in the monitor's USB, or your calibrator, do that now then re-start the software.
We also choose Advanced Mode on this screen as we of course want to get the most out of our excellent monitor hardware.
To actually being the process, we choose Profiling in the top left corner of this screen.
The next step is to choose the settings for calibration of the display.
RGB Primaries - BenQ monitors are wide gamut monitors, and we want so see and use all that lovely colour, so we choose Panel Native here. If you want to constrain the gamut of the monitor, or are aiming to work to a video standard, you could choose e.g. sRGB or DCI-P3 here as appropriate.
Whitepoint - The industry standard whitepoint is 6500K (AKA D65). You should probably use this if you are using a typical photo lab for your printing, or entering images into your camera club etc. Alternatively, if you're doing your own printing (or printing with us here), you will probably find a warmer whitepoint gives you a noticeably better screen to print match. We use 5800K here, which has become a de facto standard for many working in fine art print.
Luminance - For most print work, somewhere in the range of 80 to 120 cd/m2 is typically appropriate. After calibration, if you find e.g. your prints are always darker than you expect, re-calibrate and set a lower figure here (or vice versa if prints are lighter, use a higher figure here). 100 cd/m2 is a good starting point, so that's what we have set here. If you're doing video work, and you are trying to simulate TVs etc, you will probably want to use a higher figure, or even Maximum for the luminance.
Gamma - For almost all still imaging work, this should be 2.2. In the video space, different gammas are often used, such as 2.4. Set as appropriate.
Contrast Ratio - this control allows us to raise the black point of the monitor to reduce the screen contrast. This is mostly used in the context of preparing files for print - typically, a contrast ratio of around 250:1 or 200:1 is used. This constrains the monitor's contrast to a level much closer to that of print. However, if you are not working with files for print (i.e. you are delivering electronically, or working on video), you will probably want to use the full contrast available to the monitor as this will give the most visually pleasing display, and be a better match for what users on non-calibrated screens will see.
Next, we choose settings for the ICC profile itself.
Version 2 profiles have less compatibility issues and V4 profiles do not offer any practical benefit as such, thus we choose Bradford Adaptation, Version 2, and we choose the more accurate Table Based profiles here.
We want to get the most accuracy from our screen, so we choose to use the Large patch set. Calibration will take longer, but the results will be more accurate.
There are a couple of special features available on this page, worth mentioning.
You can increase the calibration accuracy for specific tones if you have a set of tones that are crucial to you (e.g. logo colours). Calibration will be biased towards accuracy of these tones. You can do this either by loading spot tones, or by choosing specific colours from an existing image.
The buttons to actually load these colours are rather hidden - you'll find them in the top left corner of the patch set picture.
In most typical cases, though, you don't add specific tones here.
(Should you want to, we have more info about this in our general guide to i1Profiler - this BenQ software is based on this more general calibration application).
Now we move on to the part of the process where something actually happens - the measurement stage.
You can now choose into which of the hardware calibration slots in your monitor's hardware you wish to load this calibration - e.g. the PV270 offers Calibration 1 and Calibration 2 slots. You can then easily switch between these modes using the buttons on the monitor itself.
In another section of the software (see below) you can load additional calibrations into these modes as needed, but the monitor itself can only store two calibrations at a time for easier switching between those two modes.
Tick uniformity (unless for some reason you want the monitor's full brightness (e.g. video work) - uniformity being on will constrain the maximum brightness of the screen - which is no issue when setting the monitor up for print work).
To actually start the measurement process, click the Start Measurement button found under the patch set, and follow the on-screen instructions for mounting your calibrator on the screen.
The process typically takes around 10 minutes but may take up to 30 minutes, so be patient.
When measurement is complete, you'll notice that each patch now displays both the desired and measured colour:
The final step is to create and save your ICC profile.
Give your profile a sensible name and make sure the System Level box is ticked, then hit Create and Save Profile.
Once the profile making process is complete you will receive a summary of the results, and you can use the buttons at the top left of the summary to see curves, and compare before and after images.
At this point your BenQ P series monitor is calibrated, and you can finish here if you like, or go back and set up more calibrations for different working scenarios.
You should now be able to enjoy gorgeous, accurate colour on your now calibrated display! We recommend you re-run the calibration every month or so.
Palette Master offers further features.
You can perform a quality assessment on your monitor and track it's change in behaviour over time. Worth pointing out that verifying a calibration with the same sensor used to make the calibration is a somewhat flawed approach (any mistake in measurement is likely to be repeated in verification). But it doesn't cause harm and can be interesting, so feel free to explore.
There is also a tool for testing the uniformity of your screen. This is often useful much later in a monitor's life when uniformity performance has degraded.
For advanced users, there is also a Calibration Adjustment section. Much like similar options in Eizo and NEC software, this allows you to tweak your calibration by eye to (in theory) improve your calibration.
In practise, as with the Eizo and NEC options, this is most useful for very specific, and often rather synthetic, scenarios - such as trying to match a visual image of a colour checker to one under a particular viewing light.
Typically, though, these adjustments can do more harm than good for general work - so use with caution!
We'll let you explore these more advanced options in your own time - none of these are required for a basic calibration process.
Below we link to an article that lists some suggested calibration targets for different scenarios, such as fine art printing or video editing work.
We have lots of other content you may find useful as a BenQ P Series owner, here are a few of particular relevance:
Or, browse all our colour accurate monitor articles:
- Bob H -
Thank you for your email. Actually I am a returning customer and a very satisfied customer into the bargain. This time I bought an Epson P800 - originally I bought an Epson 3880 from you and have been very satisfied with it. But having seen the Luminous Landscape videos promoting the P800 I simply had to have one. So I have sold the 3880 and upgraded. It arrived today... you really are efficient - I only ordered it yesterday!
So, thank you for efficient, friendly and excellent service - I teach photography at the local University of the Third age and refer my students to you with every confidence they will be well treated.